3 in 1 skin care Vacuum Spray Diamond Micro Dermabrasion skin peeling machine CV01
1. Aging and sun-damaged skin
2. Some types of acne scarring and acne
3. Altered pigmentation
4. Stretch marks
5. Fine lines
6. Enlarged and clogged pores
|Diamond Dermabrasion Machine||1|
|PVC vacuum hose8*5||1|
|Pens box & foam||1|
|Heads box & foam||1|
|Cotton Filter||1 package|
1. Push in the movable ring of copper nozzle at the hand piece, and pull out the 8*5 size vacuum hose. Then push in the black plastic ring on the vacuum hose.
2. Put a drop of cream or milk lotion on the vacuum hose nozzle, push in the movable ring and insert in the vacuum hose to the limit.
3. Insert a cotton swab into the Diamond Head, then tight up the Diamond Head on the hand piece. Remove the cotton swab after every treatment.
4. Turn on the power switch.
5. Adjust the Vacuum intensity by Vacuum Regulator according to the following reference table, the intensity displays on Vacuum gauge and check with the client for comfort.
6. Use the incline area of Hand piece to contact with the skin surface. Proceed with a slow sliding motion. Do not place the hand piece at one spot for too long, doing so will cause harm to the skin.
7. Adjust the crystal flow rate by the Vacuum Regulator slowly from minimum to maximum. For commonly use, do not set the flow rate to maximum unless it is of necessity. This is for better comfort intensity.
8. Gently stretch the skin with thumb and finger and glide the Hand-piece 2 to 3 times over the area being treated. Then cross again in different directions to prevent streaking lines as in the reference picture. Brush strokes or polishing motions may be used in different sites.
9. After treatment, brush off all remaining dead skin with a soft brush and rinse with damp sponges or damp cool towel.
10. Allow skin to dry then apply desired skin product.
11. To clean the Diamond Head, use toothbrush or cotton swab damped with a little alcohol to wipe off the dirty impurity, and then use UV light or high temperature for sterilization.
WHAT OUR CLIENTS SAY
This is a practical skin cleansing instrument. It can be combined with any brand of skin care products to carry out deep facial cleansing to achieve the effect of facial lifting and firming. Very suitable for small beauty salons or nail salon investment.
What to do after treatment？
◆ Avoid exercise that will create friction and make sure to gently pat off any sweat
or water to dry the skin under your dressing.
◆ For the first 3 days after treatment, keep the treated area bandaged with sterile
gauze and protected with a thin layer of antibiotic or healing ointment.
◆ After 3 days, keep the treated area clean and dry during the healing processIt
is recommended to wait at least six weeks after gettinga new tattoo to begin
the laser tattoo removal process.
◆ Don’t eat Spicy food within 1week
1.No blood, no pain, no inflammation and no need of anaesthesia;
2.No black print or scar be left;
3.No turgescence caused
4.You can resume your daily activity or work at once after therapy without delay or interference;
Equipment daily maintenance
◆gently, does not put the instrument placed in direct sunlight
◆regularly with a dry towel clean surface, machine parts surface dust and dirt.
◆before and after each use will probe cleaning and disinfection (wipe the surface with a 75% medical alcohol, prohibit the use of alcohol)
◆to prevent cross infection, each operation probe for personal use!
Next page >>
Spot Remover , Skin Care Needle Free Mesotherapy Machine / Equipment 50 / 60hz M120View More >
strong vacuum frquncy convrsion rf cavitation beauty machine RT06View More >
Vascular Areas 3d Magic Mirror System / Facial Skin Analyzer Beauty Machine MR03View More >
The Global Beauty Group is registered under the China Cooperation Act,2008 Business Number 91110115674295685F.