Fiber Coupled diode laser 808nm Permanent Hair Removal Machine FD08
◆Arm hair removal
◆Leg hair removal
◆Armpit hair removal
◆Lip hair removal
◆Bikini line hair removal
◆Whole body Perment hair removal
On fiber coupled diode laser machines the laser lamps are not located inside the hand piece but in the main body of the machine, so the heat is concentrated inside the core and not in the hand, cooling systems work less to disperse that heat so the lasers are more durable (50 Million Pulses, actually 5 times more than the best standard diodes).
The FCD (Fiber Coupled Diode Laser for Depilation) laser locates inside the mainframe instead of handpiece. Therefore, laser energy is transferred by fiber to the target, which significantly increase the energy efficiency& handpiece life span. Lighter handpiece make it more convenient for doctors, while powerful cooling maximizes patients' comfort.
FCD(Fiber Coupled Diode Laser)technology to aesthetic laser machine is a big breakthrough in medical aesthetic field. Due to the new technology, Fiber laser-I has a standing out working performance and perfect depilation effect. Built with high energy, efficacy and comfort as the foundation, Fiber laser-I is best ROI choice for a dependable laser hair removal system.
|Light Source||Fiber Coupled Diode Laser 808nm|
|Output Energy||1-50 J/cm2|
|Pulse width||10-400ms adjustable|
|Cooling system||Thermal electric cooling + water cooling+ Air cooling+sapphire contact cooling for laser head|
|LCD screen||8 inch 24 color multi color touch screen|
|Power requirement||AC220V±10%,50Hz, AC110V±10%,60Hz|
1.Optional power of 300 W and 1000 W,
2.No requirements on water quality, general purified water can be added into the water tank,
3.Excellent cooling effect, the temperature can reach -5℃,
4.Long life span of 50 million shots above,
5.Low damage rate, long service life, easy to maintain and uniform energy distribution.
◆Step One: Shave hairs Use the one-off hair removal knife to shave all the hairs on the treatment part so that the laser energy can pull off the hair by roots in the point of treatment.
◆Step Two: Apply cooling gel The cooling gel has the effect of cooling and refreshing. Apply them on the treatment part with the purpose of protecting the skin, which lower the discomfort feeling to the least during treatment.
◆Step Three: Do the hair removal treatment According to the selected energy, vertically slide the treating head closely on the skin with proper speed. Low energy setting with slow sliding while high energy setting with fast sliding.
◆Step Four: Cleaning Slightly scrape off cooling gel and clean with water. Then moisture the skin at once and apply some sun block. Within a week after treatment, don’t use Alkaline skin care products such as perfumed soaps while bathing.
Since the lamps are not inside this part anymore, the operator gets to manage a light, low temperature piece which contact cooling tip that rarely heats up, giving the user hours and hours of comfortable treatments and to the patients awesomely painless and comfy sessions. On top of that this technology make the beam of light more focused (colimated light) what assures the most effective treatment in the shorter time possible by eliminating the waste of energy on areas that are not in need of treatment.
WHAT OUR CLIENTS SAY
This 808nm diode laser hair removal and beauty instrument is beautiful. easy to use. The effect is very good. The hair removal time is short and the effect lasts long. Customers are satisfied.
What to do after treatment？
◆ Staying out of the sun.
◆ Lightening your skin. ..Shaving treatment area
◆ Avoiding other hair removal methods. .
◆ Avoiding blood-thinning medications. .
◆ usually up to 2-6 weeks before treatment, and use a broad-spectrum sunscreen daily.
Advantage Fiber diode laser
◆ light shooting unlimited
◆ Arbitrary cooling water
◆ No need filters
◆ Zero cost on handle maintence
Equipment daily maintenance
At least clean outer surface of the machine once a week using the wet cotton ball. The neutral cleaner can be applied. but make sure the liquid doesn’t leak into the machine.
◆Optical crystal cleaning
Make sure the optical crystal keeps cleaning. It needs to be cleaned after each treatment. When do the cleaning, first use the soft cloth scrub it clean, and then use another soft cloth scrub it with alcohol. At last leave it to be dry.
◆Adding deionized water
This machine is only allowed to add the denionized water. According to its using condition, it is better suggested to be added water once per three months.
Next page >>
Desktop white 600w fiber 808nm diode laser hair removal beauty machine FD06View More >
7 In 1 Multifunction Hydro Denmabrasion PDT LED Light Therapy Device L7View More >
Salon Use Multifunctional Tattoo Removal Hair Removal Diode Laser Pico Laser Machine DPL03View More >
The Global Beauty Group is registered under the China Cooperation Act,2008 Business Number 91110115674295685F.