3 in 1 low price 755nm 808nm 1064nm diode laser hair removal machine Y9C
1. Professional 808nm diode laser, suitable for hair with any color;
2. Suitable for any kinds of skin;
3. Safe and fast. It needs 3-4 treatments, the interval of each treatment are 3 weeks;
4. No harm, no side effective;
5. High enegery density output: make sure good effective for any coloring matter--120J/CM2;
6. High frequency output: make sure heating hair follicle, lasting hair removal, fast hair remvoal-10HZ;
7. Big spot size: fast and effective results--12*20mm;
8. strong cooling: instance cooling skin, safe and comfortable--5 degree.
The effective penetration depth of 808nm can reach the target tissue (hair papilla). The appropriate pulse duration ensures sufficient thermal damage to the target tissue and the surrounding tissues are almost unaffected.The right amount of energy density ensures sufficient energy output in the right time to damage the target tissue and the normal tissue is almost unaffected;The appropriate epidermal protection measures can ensure sufficient damage to the target tissue, while the epidermis is almost impervious, thus ensuring the safety of treatment.To achieve the above design requirements at the same time, the semiconductor laser hair removal series with specially designed multiple pulse laser in selecting the mode of low energy density and heat the hair follicles to 48 ℃ by sliding for the treatment of hand tools to maintain a period of time (10 HZ ). Hair follicles and growth stem cells will lose activity, so as to achieve the goal of permanent hair removal.
12 x 12 mm2
10-1,400 ms adjustable
Laser Energy density
1-140 J / cm2 adjustable
output pulse Frequency
Temperature of probe
T/T or Western Union
standard export case
220V / 110V,16A,50/60Hz
◆Step One: Shave hairs Use the one-off hair removal knife to shave all the hairs on the treatment part so that the laser energy can pull off the hair by roots in the point of treatment.
◆Step Two: Apply cooling gel The cooling gel has the effect of cooling and refreshing. Apply them on the treatment part with the purpose of protecting the skin, which lower the discomfort feeling to the least during treatment.
◆Step Three: Do the hair removal treatment According to the selected energy, vertically slide the treating head closely on the skin with proper speed. Low energy setting with slow sliding while high energy setting with fast sliding.
◆Step Four: Cleaning Slightly scrape off cooling gel and clean with water. Then moisture the skin at once and apply some sun block. Within a week after treatment, don’t use Alkaline skin care products such as perfumed soaps while bathing.
As we all know
Side effects were recorded immediately after treatment such as pain, itching, burning, erythema, and swelling. The patient was also asked about side effect from previous treatment, long term side effect such as hypo or hyper-pigmentation and scar.
The results are expressed as numbers, percent, range and average ± SD. The data were analyzed using excel program taking two tailed, “t” test, the probability level ≤ 0.05 as the lowest limit of significant difference between percentages and simple correlation test has also been calculated.
WHAT OUR CLIENTS SAY
It High-energy 808nm diode laser hair removal instrument without damaging large spots. The effect is very good. And his cooling system is very efficient. Can be used for a long time.
What to do after treatment？
◆ Staying out of the sun.
◆ Lightening your skin. ..Shaving treatment area
◆ Avoiding other hair removal methods. .
◆ Avoiding blood-thinning medications. .
◆ usually up to 2-6 weeks before treatment, and use a broad-spectrum sunscreen daily.
◆ The diode laser may do harm to eyes or cause flaming and burning. When using, some necessary
safeguard procedures are needed.
◆ Excess light energy to the treatment part may lead to heat damage on the skin.
◆ Must not look directly into the light transmitted from the treating head.
◆ Must not launch light to air. When not use, treating head should be placed on its shelf and it should
point to the treatment part during treatment.
Equipment daily maintenance
At least clean outer surface of the machine once a week using the wet cotton ball. The neutral cleaner can be applied. but make sure the liquid doesn’t leak into the machine.
◆Optical crystal cleaning
Make sure the optical crystal keeps cleaning. It needs to be cleaned after each treatment. When do the cleaning, first use the soft cloth scrub it clean, and then use another soft cloth scrub it with alcohol. At last leave it to be dry.
◆Adding deionized water
This machine is only allowed to add the denionized water. According to its using condition, it is better suggested to be added water once per three months.
Next page >>
Portable 808nm Diode Laser Permenant Hair Removal Machine Y8AView More >
980NM Skin Rejuvenation Machine 30 - 300ms Pulse Width 1 Year Warranty RBS07View More >
Medical laser blood vessel removal 980nm Diode laser removal machine RBS03View More >
The Global Beauty Group is registered under the China Cooperation Act,2008 Business Number 91110115674295685F.